The following reset can be performed to correct issues of the sensor not connecting to the display
Symptoms of the sensor not connecting to the display could include:
- No outdoor temperature (showing dashes).
- Outdoor temperature that does not change.
- No rainfall measurement.
- Inaccurate rainfall measurement.
- No wind speed.
- Inaccurate wind speed.
- Outdoor temperature is -40 F.
You would want to ensure the following first:
- Verify that the A-B-C switch in the battery compartments of both the display and sensor are set to the same letter. You may choose A, B or C; but both units must match to sync up.
- Relocate the display unit and/or the 5-in-1 sensor. Under optimal conditions, the units can be located up to 330 ft (100 m) of each other.
- Units should be placed at least 3 feet (.9 m) away from electronics that may interfere with wireless communication (such as TVs, microwaves, computers, etc).
- Use standard alkaline batteries (or lithium batteries in the sensor when the temperature is below -4ºF/-20ºC). Do not use heavy duty or rechargeable batteries. NOTE: It may take approximately 5 minutes for the display unit and sensor to synchronize after batteries are replaced.
- Bring both the sensor and display unit together and power down by removing at least one battery from each and/or removing the power cord.
- Change the A-B-C switch on both units to match on the same channel. Changing to a different channel can be helpful.
- Reinstall the battery in the outdoor sensor.
- Reinstall battery and/or power cord into the display unit.
- Let the units sit within a couple feet of each other for approximately 5 minutes or until you see a solid sensor signal.
If you still experience the issue, move forward with the following applicable troubleshooting.