If your AcuRite Iris™ (5-in-1) display unit or My AcuRite® account are reporting inaccurate wind speed readings, please review the suggestions below. These steps can also be used if your AcuRite Iris™ (5-in-1) sensor is reporting no wind speed readings at all.
If the wind speed readings are inaccurate:
- What is wind reading being compared to? Pro weather stations are typically mounted at 33-feet or higher. Make sure to compare data using a sensor positioned at the same mounting height.
- Check the location of the sensor. Ensure it’s mounted a minimum of 5-feet in the air with no obstructions around it (within several feet). The National Weather Service recommends mounting the sensor 33 feet above the ground with no obstructions within 100 feet for the most accurate wind speeds. We understand not all of our customers can mount this high. You will receive an accurate reading for the location your sensor is mounted including all surroundings. If you are comparing to a local news station, they are following the National Weather Service guidelines.
- Ensure wind cups are spinning freely. If they hesitate or stop, try lubricating with graphite powder or spray lubricant.
1. Remove the screw located below the sensor’s wind cups. Keep the screw for step 4.
2. Spray a small amount of lubricant into the hole. (WD-40 or graphite powder)
3. Tip the AcuRite Iris™ (5-in-1) Sensor upside down (so wind cups are on the bottom), then right side up several times.
4. Reinsert the screw (removed during step 1) into the hole.
If the AcuRite Iris™ (5-in-1) sensor is not reporting any wind speed, please perform the hard reset linked below:
If you are using a High Definition display unit such as model #06058M or #06088M, please use the following reset linked below:
If you believe your Wind Direction readings to be inaccurate, please review the link below before attempting the above troubleshooting.