Good evening, I just purchase the Myacurite weather station today. I have added the sensors to the myacurite website and the temperature is reading different than the unit. It is reading 137 and the unit is reading 68. Thank you, Christopher
On your five in one weather station you have the temperature calibrated to add 68 degrees, and the humidity to add 72%. I have remove the calibrations. It should bring your readings back to normal. Please let us know if there is anything further we can assist you with.